Account#This number is stamped on each double pocket envelope. If you are testing more than one sterilizer, each will have a different number to assure proper record-keeping.If you buy a new sterilizer, call us before using any spore tests. Storage of Tests: Store all double pocket envelopes in a cool dry place away from your sterilizer or other heat sources.Multiparameter internal indicators are only available for steam sterilizers (i.e., autoclaves).
Step 4: You will then receive two emails, one with your profile username and one with your profile password.
Do not open the CONTROL side of the double pocket envelope. For dry heat sterilizers, place the spore test pack or tray on the top shelf in the center or in the center of single shelf units.
For all other types of sterilizers, place the test pack or tray in a central location among other packs or trays. Process the pack or tray containing the spore packets through a normal sterilizer cycle.4.
An internal chemical indicator should be placed in every sterilization package to ensure the sterilization agent has penetrated the packaging material and actually reached the instruments inside.
An external indicator should be used when the internal indicator cannot be seen from outside the package.